This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||21 June 2012|
|PDF File Size:||16.68 Mb|
|ePub File Size:||2.79 Mb|
|Price:||Free* [*Free Regsitration Required]|
Vis spectrophotometer for hach and hach lange water analysis methods 24 pages. Page 50 Maintenance a. Drain nach and clean the body and bubble trap.
The controller contains three relays designed for use with high voltage Danger: Seat insulation against connector with no bare wire exposed.
Table of Contents Add to my manuals Add. In the United States: Please contact the Hach Service Center serving your location. Page 20 Installation 3.
The frequency will depend on the nature and concentration of dissolved and suspended solids in the sample. Allow at least mm Refer to Figure 29 and perform the following steps to replace the fuses: The data logs store the measurement data at selected intervals.
Restore sample flow to the instrument. Failure to do so could result in serious injury to the operator or damage to the equipment.
After all plumbing and electrical connections have been completed and checked, place the head on the body and supply power to the system.
Turbidimeter Installation Information Installation 3. Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair. Twist the bulb in a bach direction, pulling out slightly, until it is released from the housing. The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value.
Hach-Lange SC 100 User Manual
Wire gauge less than 18 AWG is not recommended. Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb. Section 11 Limited Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Section 1 Specifications Specifications are subject to change without notice. See Replacement Parts and Accessories on page 55 for ordering information.
Hach-Lange SC User Manual | 58 pages
Table 1 Turbidimeter Specifications Nephelometric light scatter at 90 degrees relative to the incident light beam. Installation Electrical Installation High-voltage wiring for sc100 controller is conducted behind the high voltage DANGER The instrument must be installed barrier in the controller enclosure. Pay attention to all danger and caution statements.
System Startup Two types of verifications are offered. Section 5 System Startup 5.
Hach sc100 1720E Instrument Manual
The turbidimeter sensor must be hacb within six feet of the controller unless an extension cable is used. Rinse with deionized water before calibrating. The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk. Don’t have an account? Remove the quick s1c00 fitting and wires and install the threaded plug on the opening to maintain the environmental rating. Section 7 Troubleshooting 7.
HACH SC E INSTRUMENT MANUAL Pdf Download.
To print the manual completely, please, download it. The time is available only in hour military format. Tug gently to check the connection. Replacing The Lamp Assembly jach Page 39 Operation 4. Wet Verification System Startup 5. Section 4 Operation 4.
Index Index Numerics 90 Degree Detector Disconnect and remove the existing wires between the quick connect and terminal block J5, see Figure 21 on page 4.